|Oxygen Adjustment:||Manual||Oxygen Range:||21%- 100%|
|Low Flow Mode:||2- 25L/min||High Flow Mode:||10- 60L/min|
|Temperature Settings:||31℃, 34℃,37℃||Screen:||4.3 Inches|
|Trend Review:||1、3、7 Days Review||Dimension:||340*228*162mm|
3days Trend review High Flow Nasal Cannula Device,
High Flow Nasal Cannula Device For ICU,
high flow nasal oxygen machine for ICU
Micomme HFNC device 60A
A typical HFNC system consists of a flow generator, active heated humidifier, single-limb heated circuit, and nasal cannula.
High flow nasal cannula is a heated and humidified system that allows prescribed fraction of inspired oxygen (FIO2) levels to be delivered at very high flow rates.
Acute hypoxic respiratory failure (ARF) was the first condition treated with HFNC in adults. The use of heated and humidified high flow nasal cannula (HFNC) has become increasingly popular in the treatment of patients with acute respiratory failure through all age groups.
The main physiological effects of HFNC are maintenance of a known stable alveolar FiO2, optimal humidification of the airway mucosa, reduction of the dead space, generation of a positive end-expiratory pressure (PEEP), ultimately yielding a reduction in the respiratory drive and in the work of breathing.
During the pandemic, micomme provided over 7000 units of devices globally.
At present, micomme occupies over 50% market share in high flow filed in China.
Temperature can be adjusted precisely.
The machine uses timing function, which is convenient for nurses to charge by time.
Disposable heated tube and water chamber.
|Low flow mode||2-25L/min|
|High flow mode||10-60L/min|
|Temperature settings||31℃, 34℃,37℃|
|Real time monitoring parameters||Flow, FiO2, Temperature, Treatment time|
|Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
|Auto reminder of maintenance||Yes|
|Main device disinfection|
No need disinfection for circuits system
|Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm, temperature alarm, power down alarm, low ambient temperature alarm, FiO2 not reaching preset value alarm, flow not reaching preset value alarm, reaching present treatment time alarm|