|Oxygen Adjustment:||Manual||Oxygen Range:||21%- 100%|
|Low Flow Mode:||2- 25L/min||High Flow Mode:||10- 60L/min|
|Temperature Settings:||31℃,34℃,37℃||Screen:||4.3 Inches|
|Trend Review:||1、3、7 Days Review||Dimension:||340*228*162mm|
Micomme High Flow Oxygen Therapy Machine,
therapeutic Tool High Flow Oxygen Therapy Machine,
therapeutic Tool high flow oxygen therapy equipment
Micomme heated humidified nasal cannula oxygen therapy device 60A
High Flow Nasal Cannula is a heated and humidified system that allows prescribed fraction of inspired oxygen (FIO2) levels to be delivered at very high flow rates.
Humidified high flow nasal cannula therapy is a form of non-invasive respiratory support which has been compared to continuous positive airway pressure (CPAP).
HFNC has been used as a prophylactic tool or as a treatment device mostly in patients with acute hypoxemic respiratory failure such as pre-oxygenation before intubation, immunocompromised patients and acute heart failure.
When HFNC is applied, the constant high flow of oxygen provides a washout of the anatomical dead space of the oropharynx and proximal tracheobronchial tree, which results in more efficient breathing.
In the battle against Covid-19 in Wuhan in 2020, 70% high-flow device was from micomme.
At present, micomme occupies over 50% market share in high flow filed in China.
Intelligent water-controlling system, automatically maintaining the optimal water level in the water chamber.
The machine uses timing function, which is convenient for nurses to charge by time.
||Low flow mode||2-25L/min|
|High flow mode||10-60L/min|
|Real time monitoring parameters||Flow, FiO2, Temperature, Treatment time|
||Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
||Auto reminder of maintenance||Yes|
|Main device disinfection||
No need disinfection for circuits system
||Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm, temperature alarm, power down alarm, low ambient temperature alarm, FiO2 not reaching preset value alarm, flow not reaching preset value alarm, reaching present treatment time alarm|