|Oxygen Ajustment:||Manual||Oxygen Range:||21%~100%|
|Trend Review:||1,3,7 Days|
Micomme High Flow O2 Devices,
Micomme High Flow Nasal Cannula System,
Home Care High Flow Nasal Cannula System
Micomme high flow non-invasive respiratory therapy device 60A for home
HFNC (often referred to as high-flow) systems are broadly defined as systems that provide an oxygen-gas mixture at flows that meet or exceed a patient’s spontaneous inspiratory effort.
High-flow nasal cannula (HFNC) therapy is an oxygen supply system capable of delivering up to 100% humidified and heated oxygen at a high flow rate per minute.
Acute hypoxic respiratory failure (ARF) was the first condition treated with HFNC in adults. The use of heated and humidified high flow nasal cannula (HFNC) has become increasingly popular in the treatment of patients with acute respiratory failure through all age groups.
Other applications of HFNC include the prevention of postextubation hypoxaemia, pre-oxygenation before sedation/anaesthesia and oxygenation during bronchoscopy.
The first company to research, develop and produce high-flow in China.
At present, micomme occupies over 50% market share in high flow filed in China.
Unique one-way airway design, no need for disinfection, reducing equipment turnover time and the workload of medical staff.
The machine uses timing function, which is convenient for nurses to charge by time.
|Oxygen Adjustment Setting||Manual|
|Oxygen Adjustment Range||21%- 100%|
|Low flow mode||2- 25L/min|
|High flow mode||10- 60L/min|
|Temperature settings||31℃, 34℃,37℃|
|Real time display parameters||Preset Flow, Preset FiO2, Preset Temperature, Treatment time|
|Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
|Auto reminder of maintenance||Yes|
|Main device disinfection||No need disinfection for circuits system
Wipe disinfection for device surface
|Alarm||Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm,
temperature alarm, power down alarm, low ambient temperature alarm,
FiO2 not reaching preset value alarm, flow not reaching preset value
alarm, reaching preset treatment time alarm