|Oxygen Ajustment:||Automatic||Oxygen Range:||21%~100%|
|Trend Review:||1,3,7 Days|
OH Series High Flow Oxygen Therapy Machine,
3Days Review High Flow Oxygen Therapy Machine,
OH Series high flow oxygen delivery devices
Efficient OH Series high flow nasal cannula oxygen therapy device 70C
HFNC (often referred to as high-flow) systems are broadly defined as systems that provide an oxygen-gas mixture at flows that meet or exceed a patient’s spontaneous inspiratory effort.
High-flow nasal cannula (HFNC) therapy is an oxygen supply system capable of delivering up to 100% humidified and heated oxygen at a high flow rate per minute.
HFNC is a promising novel oxygen delivery device, whose mechanisms of action offer some beneficial effects over conventional oxygen systems. It is considered to have a number of physiological effects: it improves oxygenation, dynamic lung compliance, homogeneity and end expiratory lung volume; it decreases anatomical dead space and generates a positive airway pressure that can reduce respiratory rate, the work of breathing, and enhance patient comfort.
Its use has been widely explored, particularly in ICUs and emergency departments (EDs)
HFNC also exerts its effect by providing some variable positive pressure. By providing variable positive pressure, use of high-flow nasal cannula (HFNC) increases functional residual capacity (FRC) by about 25% compared to use of traditional nasal cannula.
In the battle against Covid-19 in Wuhan in 2020, 70% high-flow device was from micomme.
At present, micomme occupies over 50% market share in high flow filed in China.
Automatic O2 concentration control technology
The machine uses timing function, which is convenient for nurses to charge by time.
Disposable heated tube and water chamber.
|Oxygen Adjustment Setting||Automatic|
|Oxygen Adjustment Range||21%- 100%|
|Low flow mode||2- 25L/min|
|High flow mode||10- 70L/min|
|Temperature settings||31℃- 37℃|
|Real time display parameters||Preset Flow, Preset FiO2, Preset Temperature, Treatment time|
|Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
|Auto reminder of maintenance||Yes|
|Main device disinfection||No need disinfection for circuits system
Wipe disinfection for device surface
|Alarm||Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm,
temperature alarm, power down alarm, low ambient temperature alarm,
FiO2 not reaching preset value alarm, flow not reaching preset value
alarm, reaching preset treatment time alarm