|Oxygen Adjustment:||Automatic||Oxygen Range:||21%- 100%|
|Low Flow Mode:||2- 25L/min||High Flow Mode:||10- 70L/min|
|Temperature Settings:||31℃-37℃||Screen:||4.3 Inches|
|Trend Review:||1、3、7 Days Review||Dimension:||340*228*162mm|
TUV Certificated High Flow Oxygen Therapy Machine,
TUV Certificated high flow oxygen therapy equipment,
Adjustable temperature High Flow Oxygen Therapy Machine
Micomme medical OH Series heated humidified nazal cannula oxygen therapy device 70C
HFNC (often referred to as high-flow) systems are broadly defined as systems that provide an oxygen-gas mixture at flows that meet or exceed a patient’s spontaneous inspiratory effort.
High-flow nasal cannula (HFNC) therapy is an oxygen supply system capable of delivering up to 100% humidified and heated oxygen at a high flow rate per minute.
High Flow Nasal Cannula (HFNC), isn’t just a standard nasal cannula cranked up to very high flow rates. It actually takes gas and can heat it to 37℃ with a 100% relative humidity and can deliver 0.21~1.00% fio2 at flow rates of up to 70 liters/min. The flow rate and fio2 can be independently titrated based on your patient’s flow and fi02 requirements.
Acute hypoxic respiratory failure (ARF) was the first condition treated with HFNC in adults. The use of heated and humidified high flow nasal cannula (HFNC) has become increasingly popular in the treatment of patients with acute respiratory failure through all age groups.
The use of HFNC in patients with Post Surgical Respiratory Failure.
Other applications of HFNC include the prevention of postextubation hypoxaemia, pre-oxygenation before sedation/anaesthesia and oxygenation during bronchoscopy.
The first company to research, develop and produce high-flow in China.
At present, micomme occupies over 50% market share in high flow filed in China.
Intelligent temperature and humidity control system
Intelligent water-controlling system, automatically refills and maintains the optimal water level in the water chamber.
The machine uses timing function, which is convenient for nurses to charge by time.
||Low flow mode||2-25L/min|
|High flow mode||10-70L/min|
|Real time monitoring parameters||Flow, FiO2, Temperature, Treatment time|
||Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
||Auto reminder of maintenance||Yes|
|Main device disinfection||
No need disinfection for circuits system
||Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm, temperature alarm, power down alarm, low ambient temperature alarm, FiO2 not reaching preset value alarm, flow not reaching preset value alarm, reaching present treatment time alarm|