|Oxygen Ajustment:||Manual||Oxygen Range:||21%~100%|
|Trend Review:||1,3,7 Days|
HFNC Nasal High Flow Machine,
HFNC high flow oxygen machine,
60L/min Nasal High Flow Machine
Micomme high flow nasal cannula HFNC OH-60A oxygen therapy device
HFNC is a simple system with clinical effects mainly dependent on flow, oxygen concentration, and temperature setting.
High Flow Nasal Cannula (HFNC), isn’t just a standard nasal cannula cranked up to very high flow rates. It actually takes gas and can heat it to 37℃ with a 100% relative humidity and can deliver 0.21~1.00% fio2 at flow rates of up to 70 liters/min. The flow rate and fio2 can be independently titrated based on your patient’s flow and fi02 requirements.
HFNC has been used as a prophylactic tool or as a treatment device mostly in patients with acute hypoxemic respiratory failure such as pre-oxygenation before intubation, immunocompromised patients and acute heart failure.
Its use has been widely explored, particularly in ICUs and emergency departments (EDs)
The first company to research, develop and produce high-flow in China.
In 2020, micomme ranked the sixth in export of medical respiratory devices in China.
Built-in oxygen-gas mixture to accurately control oxygen and adjust oxygen concentration.
Intelligent water-controlling system, automatically maintaining the optimal water level in the water chamber.
|Oxygen Adjustment Setting||Manual|
|Oxygen Adjustment Range||21%- 100%|
|Low flow mode||2- 25L/min|
|High flow mode||10- 60L/min|
|Real time display parameters||Preset Flow, Preset FiO2, Preset Temperature, Treatment time|
|Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
|Auto reminder of maintenance||Yes|
|Main device disinfection||No need disinfection for circuits system
Wipe disinfection for device surface
|Alarm||Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm,
temperature alarm, power down alarm, low ambient temperature alarm,
FiO2 not reaching preset value alarm, flow not reaching preset value
alarm, reaching preset treatment time alarm