|Oxygen Ajustment:||Automatic||Oxygen Range:||21%~100%|
|Trend Review:||1,3,7 Days|
Userfriendly High Flow Nasal Oxygen Machine,
60L/Min High Flow O2 Devices,
Userfriendly High Flow O2 Devices
User-friendly high flow nasal cannula oxygen therapy device 70B
Humidified high flow nasal cannula therapy is a form of non-invasive respiratory support which has been compared to continuous positive airway pressure (CPAP).
HFNC (often referred to as high-flow) systems are broadly defined as systems that provide an oxygen-gas mixture at flows that meet or exceed a patient’s spontaneous inspiratory effort.
Acute hypoxic respiratory failure (ARF) was the first condition treated with HFNC in adults. The use of heated and humidified high flow nasal cannula (HFNC) has become increasingly popular in the treatment of patients with acute respiratory failure through all age groups.
The use of HFNC in patients with Hypercapnic Respiratory Failure, COPD.
During the pandemic, micomme provided over 7000 units of devices globally.
At present, micomme occupies over 50% market share in high flow filed in China.
Built-in oxygen-gas mixture to accurately control oxygen and adjust oxygen concentration.
Temperature can be adjusted precisely.
Disposable heated tube and water chamber.
|Oxygen Adjustment Setting||Automatic|
|Oxygen Adjustment Range||21%- 100%|
|Low flow mode||2- 25L/min|
|High flow mode||10- 70L/min|
|Temperature settings||31℃, 34℃,37℃|
|Real time display parameters||Preset Flow, Preset FiO2, Preset Temperature, Treatment time|
|Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
|Auto reminder of maintenance||Yes|
|Main device disinfection||No need disinfection for circuits system
Wipe disinfection for device surface
|Alarm||Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm,
temperature alarm, power down alarm, low ambient temperature alarm,
FiO2 not reaching preset value alarm, flow not reaching preset value
alarm, reaching preset treatment time alarm