|Oxygen Adjustment:||Automatic||Oxygen Range:||21%- 100%|
|Low Flow Mode:||2- 25L/min||High Flow Mode:||10- 70L/min|
|Temperature Settings:||31℃,34℃,37℃||Screen:||4.3 Inches|
|Trend Review:||1、3、7 Days Review||Dimension:||340*228*162mm|
60L/Min High Flow Breathing Machine,
Micomme Medical Hfnc Oxygen Machine,
60L/Min Hfnc Oxygen Machine
Micomme medical high flow device with oxygen therapy 70B
A typical HFNC system consists of a flow generator, active heated humidifier, single-limb heated circuit, and nasal cannula.
HFNC is a simple system with clinical effects mainly dependent on flow, oxygen concentration, and temperature setting.
Acute hypoxic respiratory failure (ARF) was the first condition treated with HFNC in adults. The use of heated and humidified high flow nasal cannula (HFNC) has become increasingly popular in the treatment of patients with acute respiratory failure through all age groups.
In physiological terms, HFNO improves the fraction of inspired oxygen, washes and reduces dead space, generates positive end-expiratory pressure (PEEP) and provides more comfort than cold and dry oxygen.
In the battle against Covid-19 in Wuhan in 2020, 70% high-flow device was from micomme.
During the pandemic, micomme provided over 7000 units of devices globally.
Built-in oxygen-gas mixture to accurately control oxygen and adjust oxygen concentration.
Intelligent water-controlling system, automatically maintaining the optimal water level in the water chamber.
The machine uses timing function, which is convenient for nurses to charge by time.
|Low flow mode||2-25L/min|
|High flow mode||10-70L/min|
|Real time monitoring parameters||Flow, FiO2, Temperature, Treatment time|
|Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
|Auto reminder of maintenance||Yes|
|Main device disinfection|
No need disinfection for circuits system
|Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm, temperature alarm, power down alarm, low ambient temperature alarm, FiO2 not reaching preset value alarm, flow not reaching preset value alarm, reaching present treatment time alarm|