|Oxygen Adjustment:||Manual||Oxygen Range:||21%- 100%|
|Low Flow Mode:||2- 25L/min||High Flow Mode:||10- 60L/min|
|Temperature Settings:||31℃,34℃,37℃||Screen:||4.3 Inches|
|Trend Review:||1、3、7 Days Review||Dimension:||340*228*162mm|
Oxygen Therapy High Flow Nasal Cannula System,
High Flow Nasal Cannula System For Respiratory Dept.,
Oxygen Therapy high flow o2 devices
Medical high flow nasal cannula device with oxygen therapy 60A
High-flow nasal cannula (HFNC) therapy is an oxygen supply system capable of delivering up to 100% humidified and heated oxygen at a high flow rate per minute.
HFNC is a promising novel oxygen delivery device, whose mechanisms of action offer some beneficial effects over conventional oxygen systems. It is considered to have a number of physiological effects: it improves oxygenation, dynamic lung compliance, homogeneity and end expiratory lung volume; it decreases anatomical dead space and generates a positive airway pressure that can reduce respiratory rate, the work of breathing, and enhance patient comfort.
Acute hypoxic respiratory failure (ARF) was the first condition treated with HFNC in adults. The use of heated and humidified high flow nasal cannula (HFNC) has become increasingly popular in the treatment of patients with acute respiratory failure through all age groups.
HFNC also exerts its effect by providing some variable positive pressure. By providing variable positive pressure, use of high-flow nasal cannula (HFNC) increases functional residual capacity (FRC) by about 25% compared to use of traditional nasal cannula.
The first company to research, develop and produce high-flow in China.
At present, micomme occupies over 50% market share in high flow filed in China.
Automatic O2 concentration control technology
Effectively reduce the dead space with 60L/min high flow
The machine uses timing function, which is convenient for nurses to charge by time.
||Low flow mode||2-25L/min|
|High flow mode||10-60L/min|
|Real time monitoring parameters||Flow, FiO2, Temperature, Treatment time|
||Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
||Auto reminder of maintenance||Yes|
|Main device disinfection||
No need disinfection for circuits system
||Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm, temperature alarm, power down alarm, low ambient temperature alarm, FiO2 not reaching preset value alarm, flow not reaching preset value alarm, reaching present treatment time alarm|