|Oxygen Ajustment:||Automatic||Oxygen Range:||21%~100%|
|Trend Review:||1,3,7 Days|
High Efficiency High Flow Nasal Cannula Device,
High Efficiency high flow oxygen equipment,
3days Review High Flow Nasal Cannula Device
High Efficiency 3days Review High Flow Nasal Cannula Device
HFNC (often referred to as high-flow) systems are broadly defined as systems that provide an oxygen-gas mixture at flows that meet or exceed a patient’s spontaneous inspiratory effort.
High flow nasal cannula is a heated and humidified system that allows prescribed fraction of inspired oxygen (FIO2) levels to be delivered at very high flow rates.
Acute hypoxic respiratory failure (ARF) was the first condition treated with HFNC in adults. The use of heated and humidified high flow nasal cannula (HFNC) has become increasingly popular in the treatment of patients with acute respiratory failure through all age groups.
The use of HFNC in patients with Post Surgical Respiratory Failure.
The first company to research, develop and produce high-flow in China.
In the battle against Covid-19 in Wuhan in 2020, 70% high-flow device was from micomme.
Revolutionary trend review function, review the latest 1,3,7 days data of temperature, humidity and O2 concentration.
The machine uses timing function, which is convenient for nurses to charge by time.
Disposable heated tube and water chamber.
|Oxygen Adjustment Setting||Automatic|
|Oxygen Adjustment Range||21%- 100%|
|Low flow mode||2- 25L/min|
|High flow mode||10- 70L/min|
|Temperature settings||31℃- 37℃|
|Real time display parameters||Preset Flow, Preset FiO2, Preset Temperature, Treatment time|
|Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
|Auto reminder of maintenance||Yes|
|Main device disinfection||No need disinfection for circuits system
Wipe disinfection for device surface
|Alarm||Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm,
temperature alarm, power down alarm, low ambient temperature alarm,
FiO2 not reaching preset value alarm, flow not reaching preset value
alarm, reaching preset treatment time alarm