|Oxygen Adjustment:||Manual||Oxygen Range:||21%- 100%|
|Low Flow Mode:||2- 25L/min||High Flow Mode:||10- 60L/min|
|Temperature Settings:||31℃，34℃，37℃||Screen:||4.3 Inches|
|Trend Review:||1、3、7 Days Review||Dimension:||340*228*162mm|
Home Use Hfnc Devices,
Manual High Flow Therapy Machine,
Home Use High Flow Therapy Machine
Heated humidified high flow nasal cannula oxygen therapy device (HFNC)
Humidified high flow nasal cannula therapy is a form of non-invasive respiratory support which has been compared to continuous positive airway pressure (CPAP).
High Flow Nasal Cannula (HFNC) isn’t just a standard nasal cannula cranked up to very high flow rates. It actually takes gas and can heat it to 37℃ with a 100% relative humidity and can deliver 0.21~1.00% fio2 at flow rates of up to 60 liters/min. The flow rate and fio2 can be independently titrated based on your patient’s flow and fi02 requirements.
HFNC has been used as a prophylactic tool or as a treatment device mostly in patients with acute hypoxemic respiratory failure such as pre-oxygenation before intubation, immunocompromised patients and acute heart failure. When HFNC is used to deliver oxygen, the flow rates are much higher than can be achieved with traditional nasal cannula. This results in a greater delivery of prescribed oxygen into the lungs, and less entrainment of room air. The oxygen you want to deliver to your patients is not prone to the same effect of dilution!
Micomme is the first company to research, develop and produce high-flow in China.
During the pandemic, micomme provided over 7000
pieces units of devices globally.
Automatic O2 concentration control technology
Effectively reduce the dead space with 60L/min high flow
Intelligent water-controlling system, automatically refills and maintains the optimal water level in the water chamber.
|Low flow mode||2-25L/min|
|High flow mode||10-60L/min|
|Temperature settings||31℃, 34℃, 37℃|
|Real time monitoring parameters||Flow, FiO2, Temperature, Treatment time|
|Trend review||1,3,7 days review|
|Content of review||Flow, FiO2, Temperature, Treatment time|
|Presetting of treatment time||Yes|
|Auto reminder of maintenance||Yes|
|Main device disinfection||
No need disinfection for circuits system
Wipe disinfection for device surface
|Tube alarm, oxygen pressure alarm, blocking alarm, water level alarm, temperature alarm, power down alarm, low ambient temperature alarm, FiO2 not reaching preset value alarm, flow not reaching preset value alarm, reaching present treatment time alarm|